EXERCISES, LESSON 3
INSTRUCTIONS: Answer the following exercises by circling the lettered response that
best answers the question.
After you have answered all of the exercises, turn to "Solutions to Exercises" at
the end of the lesson and check your answers. For each exercise answered incorrectly,
reread the lesson material referenced after the solution.
Which of the following procedures do you use to calibrate the developer scale
offset on the AFP 14-X3?
a. For access to the developer tank, remove the side panels.
b. Place an accurate mercury process thermometer in the developer solution.
Note the voltage between TP3 and TP1 (common) on J3 which should be
d. Set temperature dial to actual solution temperature and adjust the voltage
back to the previously noted voltage.
Which of the following procedures do you use to calibrate the replenishment on
the AFP 14-X3?
a. Bleed air from the replenisher lines and check that the pump is pumping
b. Calibrate the developer and fix replenisher pump heads at the same rate.
Adjust the rate of replenishment by turning the volume screw on the
replenisher pump clockwise to decrease the rate.
d. Always tighten down the locknut when you have completed the adjustment.