e. Record the pulses on the monitor.
f. If not within tolerance, adjust the pulse count before proceeding. To adjust
the pulse count, use the following procedures.
Make an exposure.
(2) Adjust the pulse using potentiometer R13 on circuit board D1 until the
proper pulse count is obtained. Refer to figure 2-10.
+ = Increase the exposure time.
- = Decrease the exposure time.
Figure 2-10. Pulse count adjustment.
Observe the cooling time between exposures. See the cooling chart in
the manufacturer's service manual to determine the correct cooling
period for a relative power duration.