c. Test Gauge. Use an A-DEC air pressure gauge with
a 0-100psi range for checking air pressure at various points while troubleshooting the
system. You also require the following to use this gauge: barbed fitting, washer and
sleeve clamp kit, tee barb and sleeve clamp kit, and a two-foot length of 1/8-inch tubing.
Refer to figure 6-3.
Figure 6-3. Test gauge.
d. Syringe Tool. This tool incorporates a valve core tool, a 3/32-inch hex
wrench for adjusting the syringe flow, and a 5/32-inch hex wrench for removing and
installing the syringe tip insert. Refer to figure 6-4.
Figure 6-4. Syringe tool.