(q) Radiographic and fluoroscopic kvp.
Ma meter and shunt circuit.
Ma meter circuit.
Single-phase impulse-timer circuit.
(w) Single-phase SCR-contactor circuit.
(aa) Autotimer and back-up timer interlock.
(bb) Autotimer reset and field selector.
(cc) Auxiliary-relay circuit.
(dd) Autotimer-technique selector.
a. Introduction. Maintenance for the MXT-90/15 radiographic/fluoroscopic
table falls into two basic categories: preventive and corrective. Some preventive
maintenance procedures are performed by operators. The information in this section is
intended for use by a Medical Equipment Repairer.
b. General Inspection. When inspecting the MXT-90/15 table, be sure to do
(1) Check all nuts and bolts, screws, rivets, and other fasteners to ensure
they are tight. Also ensure all fasteners are present and in good condition.