Figure 2-2. Ventilator interior.
(a) Disconnect the green nebulizer jet line from the outlet in the bottom
of the ventilator. Install a nebulizer power line test gauge (P/N 6758, 0-60psig gauge
with green tubing) into this outlet and close this system by securing the nebulizer jet line
to the female adapter of the test gauge.
(b) Check for a single-cam stop secured on the nebulization valve at
the 6 o'clock position (figure 2-3).