d. Inspect and Test the Fan.
Remove the assembly cover.
Inspect the fan for cleanliness. Clean it if necessary.
Apply power to the fan and verify that it operates.
e. Test the Heater.
Prime the pump.
(a) Toggle the switch labeled HEATER to the ON position.
(b) Move the knee-operated lever to the activated position (right). This
will supply power to the heater.
(c) Ensure that the indicator light lights to indicate the heater is
operating. Ensure the water heats to temperatures shown in the specifications.
f. Inspect the Lamp. Verify that the lamp operates when the unit is energized.
g. Inspect the Hose Assembly. Inspect all hoses for leaks and signs of
h. Inspect the Washers. Visually inspect washers for signs of deterioration.
i. Inspect the Surgical Detergent Dispenser. Visually check for cracks,
leaking, and deterioration.
j. Inspect the Lever Control. Check for looseness, binding, and proper
k. Test the Cam Micro Switch. Rotate the lever control. Verify that the pump
energizes, and the heater works when activated.