(b) Remove the three motor mounting screws and slide the motor
forward to disengage from the chain. Lift the motor and the capacitor out.
Repair or replace the chain as necessary.
Install the new gear motor and the capacitor.
(10) Reconnect all of the electrical leads in the junction box so that the motor
is ready to operate.
(11) Reconnect the external power source.
(12) Run the motor counterclockwise until it stops to correspond to the raised
condition of the lift.
On the PLR-200, the lift will not operate without the jumper cord originally
shipped with the lift. If the jumper cord is not available, you must connect the
receptacle to the chair cord. Refer to figure 3-6.
Figure 3-6. The PLA-200 precision lift.