(1) Heading. Refer to figure 2-5. You complete this section by placing an
"X" or a check mark in the appropriate block when the inspection and performance
testing have been completed. Also enter the location, serial number, index number,
material management control number (MMCN), date of inspection, manufacturer,
model, work-order number, and the next inspection due date.
Figure 2-5. Heading.
(2) Section I, visual inspection. Refer to figure 2-6. Visually inspect the
items listed in blocks 1 through 8. If no action is required, place an "X" or a check mark
"√" in the "ok" column. If action is required, annotate action needed in the column
provided (for example, replace, repair, and so forth). When the required action has
been taken, enter the date and your initials in the "action taken" column.
Figure 2-6. DA Form 5624-R, section I.
Action is required if the unit fails a test. Equipment must operate according to
the standards set out in TM 8-605. If the equipment does not operate
according to those standards, then the equipment must be repaired.