Figure 2-11. Defibrillator beeper removal.
Figure 2-12. Switch plate removal.
Turn the board over and remove the four screws holding the switch plate
Remove the switch plate.
Exercise extreme care when replacing the 590235 board. Do not crush or
force the resistors indicated in figure 2-13 out of position.