(2) Angle the collimator 0 degrees beam-down direction with a measured
SID of 40 inches measured to the Bucky. For tilting tables, angle the table to 0 degrees
The MANUAL indicator should be ON.
Light-emitting diode-2 on the logic PCB should be ON.
The 40-inch indicator should be ON.
(3) Locate a cassette in the table cassette holder and fully insert the tray
into the Bucky.
(a) The MANUAL indicator should switch OFF.
(b) The HOLD indicator should switch ON.
Light-emitting diode on the Logic PCB should switch OFF.
(d) The collimator field size indicators (front panel) should rapidly
adjust and stop.
(e) The HOLD indicator should switch OFF.
The READY indicators should switch ON.
(g) The light-field size on the table-top should be about the same size
as the cassette inserted into the cassette tray. (Do not remove the cassette at this
(4) Angle the collimator to properly aim at the wall cassette holder with the
horizontal SID at about 50 inches.
Although the cassette was left in the table cassette tray, it should not produce
an IR TRUE signal (LED-2 should be ON and the MANUAL indicator should
Fully insert a cassette into the wall cassette holder.
Light-emitting diode-2 on the logic PCB should switch OFF.
(b) The MANUAL indicator should switch OFF.
The HOLD indicator should switch ON.
(d) All SID indicators should be OFF.