Figure 1-9. Defibrillator circuit board.
Inspect all assemblies for burnt or loose components.
(5) Inspect all chassis and panel mounted components for looseness,
breakage, loose contacts, or loose conductors.
Inspect for disconnected, broken, cut, loose, or frayed wires and cables.
(7) Inspect the patient paddles for pitted conditions and excess buildup of
Inspect the patient cables for discoloration, broken insulation, and loose
e. Perform an Operational Check on the Defibrillator. After performing visual
checks and inspections, you perform the operational checks. Perform the checks daily or
at each shift change. Refer to Figure 2-2 in lesson 2 for procedures to perform daily