Figure 1-4. Defibrillator back panel.
(17) Voltage selector switch (34). Refer to figure 1-5 for a more detailed view.
Use the voltage selector switch to select proper voltage for 110v or 220v line power.
(18) Fuse holder (35). The fuse holder holds the appropriate fuse for the line
Figure 1-5. Voltage selector switch.
PERFORM A VISUAL CHECK ON DEFIBRILLATOR AND MONITOR
Now that you know the functions of the controls and status indicators, you can
perform the inspections and tests to ensure the readiness and optimum working condition
of your MRL Porta Pak 90. Be sure to follow the safety precautions when accessing the
monitor and defibrillator main circuit board.
a. General Safety Precautions. Perform the following before disassembling the
The defibrillator has 1350v.
Turn off the monitor.